# ISA Interchange

Welcome to the official blog of the International Society of Automation (ISA).

This blog covers numerous topics on industrial automation such as operations & management, continuous & batch processing, connectivity, manufacturing & machine control, and Industry 4.0.

The material and information contained on this website is for general information purposes only. ISA blog posts may be authored by ISA staff and guest authors from the automation community. Views and opinions expressed by a guest author are solely their own, and do not necessarily represent those of ISA. Posts made by guest authors have been subject to peer review.

# AD Converters Quiz: convert an analog signal to a digital one

### Subscribe Here!

AD Converters are necessary to convert an analog signal to a digital one that our computers can understand. The output of an AD Converter is a binary number. Digital instruments most commonly use 12, 14 and 16-bit converters, although there are much higher bit converters available.

The following is a practice question from one of ISA’s classes:

1. Calculate the incremental steps for a 12-bit A/D converter

2. If the sensor connected to the A/D converter was ranged for 0-1000 F, what is the smallest increment of temperature that can be seen on the digital side of the A/D converter?

3.  If I wanted to control the temperature to within 0.1 F can I do this with a 12-bit A/D converter in the sensor module?

4. What effect would a 14-bit A/D have?

5. Can you think of further improvements?

LSB represents = 100% / 2^12 = 100%/4096 = 0.0244%

LSB (or smallest increment of temperature on the digital side):

= 1000 F / 2^12 = 1000 F/ 4096 = 0.244 F

• A 12-bit A/D converter in the sensor module is not accurate enough.
• A 14-bit A/D converter would give a resolution or LSB of :
• = 1000  F / 2^14 = 1000  F / 16384 = 0.061  F
• 14-bit A/D converter would be better for control to within 0.1 F, however a 16-bit A/D converter would be even better because the resolution would be 0.0153 F.

Some points to remember are that although a higher bit converter allows for highly accurate instruments, we want our test equipment that we use to calibrate them to be at least 4 times more accurate. So for example if we had a transmitter with an accuracy of .02% FS, then our test equipment would have to have an accuracy of .005% FS. Also if the AD Converter on the host system’s analog input card (when the transmitter’s digital signal is being converted to 4-20) is not equal or higher in bits then the transmitter’s, the resolution that you gained is lost.

###### Mark Weisner
Mark Weisner is a technical specialist at ISA.